|ROSAT Home Page||ROSAT
For #1, an examination of light curves and plots of the KS test show that there are many intervals and obis when no activity is seen. #2 is quite different, being more steady (less ``sparking'' or ``flickering''). For both, the average count rate drops quickly as the length of the time bin is increased.
|HOT SPOT #1||HOT SPOT #2|
|First known occurrence||8Jul95/rh800694||mid-Mar95|
|Last known occurrence||continues (Jan96)||19Jun95|
|Largest known counts s (1 sec bins)||57(rh300399)||much less than #1|
|Largest known <counts s for a seq:||0.76 (17.2ksec)||0.0177(28 ksec)|
|Incidence of Detection||120 out of 240 seqs;||7/240|