Venera 11 & Venera 12
Veneras 11 and 12 were part of the French-Soviet collaboration SIGNE 2. The SIGNE 2 project consisted of French-built gamma ray detectors on the Soviet satellites Venera 11, Venera 12, and Prognoz 7. These 3 satellites then formed an interplanetary gamma-ray burst triangulation network. Two features made this program unique in its time: it was the first time that identical detectors had been used simultaneously for gamma-ray burst studies on separate spacecraft over interplanetary distances, and the experiments used the largest dedicated gamma-ray burst detectors put into orbit.
Both Veneras had two dedicated gamma-ray burst detectors, arranged in such a way that they viewed 4-pi steradians. Each detector consisted of a 4.5 X 3.7 cm NaI(Tl) crystal surrounded by an 8mm thick plastic anticoincidence jacket. The crystal and plastic were viewed from the side by photomultiplier tubes.
The gamma-ray burst detector operated in low time resolution "waiting" modes in the absence of a burst; low energy resolution spectra and higher energy resolution calibration spectra were transmitted in this mode. Typically, the data were accumulated into 120.0 s time bins, 1-31 energy channels. Detection of a burst triggered the storage of high time resolution count rates and spectral data. The trigger criterion was an excess count rate 8 sigma above normal in a 250 ms interval. The maximum time resolution was 1/512 s. The energy ranges examined for burst detection ranged from 50-270 keV to 600-3000 keV, depending on the detector mode. During a burst, there were 1-6 energy channels used, and, for a "typical" burst seen with Venera 11 & 12, the nominal energy range was 80-800 keV. The readout time was ~4 s. However, two successive bursts could only be detected if the interval between them was 3-20 minutes. In addition, the storage of information on-board the spacecraft limited the number of distinguishable bursts to 1 every 2 hours.
Also aboard the 2 spacecraft was the KONUS experiment. It consisted of 6 scintillation detectors, a set of 6 devices for gamma-ray burst detection against the current background, 6 counters to measure the count rate in each sensor, a 320-channel time analyzer, and a 128 channel PHA. Each scintillator was a NaI(Tl) crystal with 80 mm diameter and 30 mm thickness. The background was measured over the energy range 50-150 keV for count rate, and over the range 30 keV - 2 MeV for spectra. Eight seconds of preburst data was recorded when a trigger was issued, with a resolution of 0.25 s. The temporal profile of the burst was recorded in 3 ways: 2 s with 15.625 ms resolution, 32 s with 0.25 s resolution, and an additional 32 s with 1 s resolution. The PHA trigger mode was a 16 channel, quasi-logarithmic energy scale mode which measured 8 energy spectra in 8 successive intervals of 4 s each. The threshold sensitivity at the 6-sigma trigger level was ~4 X 10-7 erg/cm2.
HEASARC Home | Observatories | Archive | Calibration | Software | Tools | Students/Teachers/Public
Last modified: Thursday, 24-Sep-2020 17:37:05 EDT
The HEASARC is hiring! - Applications are now being accepted for a scientist with significant experience and interest in the technical aspects of astrophysics research, to work in the HEASARC at NASA Goddard Space Flight Center (GSFC) in Greenbelt, MD. Refer to the AAS Job register for full details.